We purchase Elite coriolis meters for piping leak detection purpose. The meter is installed according to the above photo link
1. Is the pipe support sufficient for the meter
2. If the is pipe stress or displacement, can this effect be zerorised during meter calibration
3. Is there guidance of max allowable nozzle load onto coriolis meter flange
4. Is there examples of successfully completed Elite flow meter in “flag” position?
We have minimal space to route the pipe and placed the support, thus this design was chosen
Thanking you in advance for your inputs
Thank you for your questions regarding our ELITE Series Coriolis meters. Please see below for our responses:
For more information, please see our ELITE Series Installation Manual here: https://www.emerson.com/documents/automation/installation-manual-elite-meters-en-64354.pdf
For any additional questions, please contact us at Flow.email@example.com.
Max flange loads updated 2017.pptx
In reply to Ayuko Morikawa:
Hi Ayuko, Thank you for your valuable reply. We understand that meter's support through pipe flange is acceptable. Based on our pipe flex check, the piping configuration will yield some displacement on Coriolis meter flange during operating condition (i.e. up to 2.5mm at top flange and 0.5mm at bottom joint from zero flow condition). The stress on meter flange is less than the allowable stress limit that you shared. Below schematic illustrates the displacement in 1D (pardon me for the poor hand dwg)
Will the displacement affect the meter's performace? To achieve zero displacement on coriolis meter flanges, independent welded support is needed next to meter flange joints. Is this necessary? Looking forward to your reply and thanking you in advance for your further advice
In reply to Tom O'Banion:
In reply to Mea05key:
This is the official online community site of the Emerson Global Users Exchange, a forum for the free exchange of non-proprietary information among the global user community of all Emerson Automation Solution's products and services. Our goal is to improve the efficiency and use of automation systems and solutions employed at members’ facilities by sharing our knowledge, experiences, and application information.
User Groups |
World Areas |
Community Guidelines |
Legal Information |
Contact Community Manager
Website translation provided by
© 2015-2019 Emerson Global Users Exchange. All rights reserved.