We use 3300HTVP pH to neutralize inclinator waste acidic liquids. Sensor is mounted directly to the process. There is no option for effluent disposal. The sensor is get damaged very frequently typically one or two weeks. I doubt some small particulates damaging the glass electrode. Operation maintain high flow for mixing chemicals. They are expecting pH reading with no lag. Kindly give a solution.
This is the official online community site of the Emerson Global Users Exchange, a forum for the free exchange of non-proprietary information among the global user community of all Emerson Automation Solution's products and services. Our goal is to improve the efficiency and use of automation systems and solutions employed at members’ facilities by sharing our knowledge, experiences, and application information.
User Groups |
World Areas |
Community Guidelines |
Legal Information |
Contact Community Manager
Website translation provided by
© 2015-2020 Emerson Global Users Exchange. All rights reserved.