Hi, I urgently need some advice on an installation where a Coriolis sensor is used for flow and density measurement.
When flow is routed through the sensor an drive is reported to 100%. Severe two phase alarm and drive overrange is reported. There is no reliable measurement of flow.
I have done several smart meter verifications and they always pass.
When I do a verify zero, it fails. Then I calibrate new zero and the meter reports zero flow. If left alone for some time, it starts showing flow when there is no flow in the line. A new verify zero fails and a new calibrate zero is needed.
The system is designed for a maximum flow of 7500 L/min, so this might be to big for the task at hand.
I tried to contact support, but in my region only a sales representative was available and he could not help. Any help regarding this installation is appreciated.
In reply to Bjarte Haaland:
This is the official online community site of the Emerson Global Users Exchange, a forum for the free exchange of non-proprietary information among the global user community of all Emerson Automation Solution's products and services. Our goal is to improve the efficiency and use of automation systems and solutions employed at members’ facilities by sharing our knowledge, experiences, and application information.
User Groups |
World Areas |
Community Guidelines |
Legal Information |
Contact Community Manager
Website translation provided by
© 2015-2019 Emerson Global Users Exchange. All rights reserved.