Typically having the analyzer not powering up could be for the following reasons..
1.- No 24VCD Present(please double check)
3.- 24 VCD Terminal not seated correctly on the receptacle.
4.- LOI Board and CPU not seated correctly.
5.- Backplane Board Damaged.
Note: Please make sure all connectors are correctly inserted.
By The Way , please provide instrument sales order and serial number of your GC370XA so I can check the factory records
Raymundo Morin | Global Training Manager | Rosemount Gas Analysis
Emerson Automation Solutions | 10241 West Little York, Suite #200 | Houston | TX | 77040 | USA
T +1 | M +1 346-773-1367 | F +1 713 396 8702 | Raymundo.Morin@emerson.com
Please guide about following,
Thanks and Regards,
In reply to musabthelion:
This is the official online community site of the Emerson Global Users Exchange, a forum for the free exchange of non-proprietary information among the global user community of all Emerson Automation Solution's products and services. Our goal is to improve the efficiency and use of automation systems and solutions employed at members’ facilities by sharing our knowledge, experiences, and application information.
User Groups |
World Areas |
Community Guidelines |
Legal Information |
Contact Community Manager
Website translation provided by
© 2015-2020 Emerson Global Users Exchange. All rights reserved.