We have AMS ver. 12.5 and the Druck DPI 620 Genii. AMS downloaded test schemes use Channel One for the current (mA) measurement. This channel has a higher impedance which means current measurement using the Rosemount test connections is inaccurate. Has anyone found a work around for this? Disconnecting the negative wire to series the calibrator into the loop is not an option for us.