We have AMS ver. 12.5 and the Druck DPI 620 Genii. AMS downloaded test schemes use Channel One for the current (mA) measurement. This channel has a higher impedance which means current measurement using the Rosemount test connections is inaccurate. Has anyone found a work around for this? Disconnecting the negative wire to series the calibrator into the loop is not an option for us.
This is the official online community site of the Emerson Global Users Exchange, a forum for the free exchange of non-proprietary information among the global user community of all Emerson Automation Solution's products and services. Our goal is to improve the efficiency and use of automation systems and solutions employed at members’ facilities by sharing our knowledge, experiences, and application information.
User Groups |
World Areas |
Community Guidelines |
Legal Information |
Contact Community Manager
Website translation provided by
© 2015-2019 Emerson Global Users Exchange. All rights reserved.